Service Guide. Publication Number (order as manual set) The security code is set to “HP” when the function generator is. Note: Unless otherwise indicated, this manual applies to all Serial Numbers. The Agilent .. For information on using the Phase-Lock Option for the HP A. Microsoft® Windows® included. Agilent A. 15 MHz Function /. Arbitrary Waveform Generator. Note: Unless otherwise indicated, this manual applies to all .
|Published (Last):||14 November 2013|
|PDF File Size:||6.32 Mb|
|ePub File Size:||1.14 Mb|
|Price:||Free* [*Free Regsitration Required]|
Chapter 3 Features and Functions System-Related Operations Display Control To speed up the rate at which the function generator can make configuration changes, or for security reasons, you may want to turn off the front-panel display. The number of frequency points in the manuxl depends on the manula time you select and is automatically calculated by the function generator. Page 88 Chapter 3 Features and Functions Frequency Sweep Sweep Trigger Source In the triggered sweep mode, the function generator outputs a single sweep each time a trigger is received.
Certain combinations of burst count and carrier frequency are not allowed. Page 11 Chapter 1 Quick Start Hl33120a adjust the carrying handle To adjust the carrying handle To adjust the position, grasp the handle by the sides and pull outward.
Page 45 Features and Functions You will find that this chapter makes it easy to look up all the details about a particular feature of the function generator. The default is Hz. Chapter 3 Features and Functions Calibration Overview To Unsecure for Calibration You can unsecure the function generator for calibration either from the front panel or over the remote interface.
This enables you to recall the entire instrument configuration with a single command from the remote interface or with just a few key presses from the front panel. Page 86 Chapter 3 Features and Functions Frequency Sweep Start Frequency and Stop Frequency The start frequency and stop frequency set the upper and lower frequency bounds for the sweep. The following steps show you how to store and recall a state. Page 26 Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial Menu Example 1 The following steps show you how to turn on the menu, move up and down between levels, move across the choices on each level, and turn off the menu.
The Trig annunciator turns on when the function generator is waiting for a single trigger. Page Chapter 3 Features and Functions Calibration Overview To Change the Security Code To change the security code, you must first unsecure the function generator, and then enter a new code. The burst rate frequency defines the interval between bursts. Recall 5 Recall the stored instrument state.
Don’t have an account? When you enable the burst mode, the previous modulation mode is turned off. Page 50 Chapter 3 Features and Functions Output Configuration Output Amplitude As shown below, the output amplitude range depends on the function currently selected and the output termination. This enables you to recall the entire instrument configuration with just a few key presses from the front panel.
Chapter 2 Front-Panel Menu Operation To turn off the comma separator To turn off the comma separator The function generator can display values on the front panel with or without a comma separator.
When the gate signal is true, the function generator outputs a continuous waveform. This is the correct fuse for all line voltages. The function generator will accept only an internal modulating signal no external source is available. Only one interface can be enabled at a time. Notice that the annunciator flashes, indicating that the displayed percentage is the depth also called percent modulation.
Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator. The function generator outputs one burst or Single initiates one sweep each time you press the key.
The rate is set to baud when the function generator is shipped from the factory. Increments the flashing digit.
The default amplitude is mVpp into 50 ohms for all functions. The HP-IB address can be set from the front-panel only. Page Chapter 3 Features and Functions System-Related Operations System-Related Operations This section gives information on topics such as power-down recall, self-test, error conditions, and mannual display control. See the next page if you need to change the power-line fuse.
Notice that the annunciator flashes, indicating that the displayed Burst parameter is for the burst mode. After the specified number of cycles has been output, the function generator waits for the next trigger while outputting no signal zero volts or the dc offset level.
Chapter 3 Features and Functions Triggering Single Triggering In the single trigger mode front panel onlyyou can manually trigger the function generator by pressing the front-panel key.
Be sure to read the security code rules on page before changing the security code. You may never have to change some of the parameters discussed here, but they are provided to give you the flexibility you might need. The front-panel display will light up while the function generator performs its power-on self-test.
Page Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator.
Be sure to read the security code rules on page before attempting to secure the function generator. To recall the stored state, you must use the same memory location used previously to store the state. However, performance test data is available upon request. The function generator beeps and displays a message to show that the change is now in effect.
Page Chapter 3 Features and Functions System-Related Operations Display Control To speed up the rate at which the function generator can make configuration changes, or for security reasons, you may want to turn off the front-panel display. Add to my manuals Add.
Inthe amplitude of the carrier is varied by the amplitude of the modulating waveform. Ampl 1 Enable the amplitude modify mode.
Use the arrow keys to edit individual digits. You can output the burst at a rate determined by the internal rate generator or the signal level on the rear-panel Ext Trig terminal. Output Minimum Maximum Function Termination Page 19 Chapter 1 Quick Start To store the instrument state To verify that the state was stored properly, you can turn the power off before recalling the state. We have written the exercises in this chapter to prepare the function generator for use and help you get familiar with some of the front-panel operations.
Display Annunciators Adrs Function generator is addressed to listen or talk over a remote interface.