View and Download Escort A operation manual online. Count Multimeter. A Multimeter pdf manual download. Here you will find a large selection of new and used Escort A listed at special prices. Although some items match several categories, all of them are listed. Section 7. GPIB Remote Operation. Introduction. Description of the GPIB. GPIB Interface Parameters Setup. Commands Summary. 72 .
|Country:||Moldova, Republic of|
|Genre:||Health and Food|
|Published (Last):||21 October 2005|
|PDF File Size:||6.84 Mb|
|ePub File Size:||10.37 Mb|
|Price:||Free* [*Free Regsitration Required]|
Section 2 Getting Started Introduction Section 2 describes the front panel operational keys, displays, input terminals and rear panel of the meter, adjusting handle, explains general operating features. As configuration data for IEEE address, RS baud rate, data bit, stop bit, parity, echo and so on are stored in escrot non-volatile memory, they are not changed when power is cycled off and on until the configurations are changed by the user.
The “trig” command may then be used to trigger a measure and get the related data. Do not apply a voltage exceeding V peak between and 3136w input terminals. Sets the display 1 as ohmeter for tests of continuity.
Then, the nearest compatible range is set. To print the manual completely, please, download it. This manual contains information and warns which must be followed to ensure safe operation and retain the meter in safe condition. The front and rear protective holsters can be removed when mount escortt meter into a rack. However, at different range, the HI and the LO limits represent different values according to their respective counts.
Please refer to Section for detail information about Query Commands. The power-up configuration set at the factory is shown in Table To display DC voltage in primary display, and AC escirt in secondary display or vise versus while testing a rectifier circuit, user may check the DC voltage supplied and its AC ripples by taking a single meter.
Introduction Section 4 Measurement Application Ewcort Introduction Section 4 describes some advanced features and applications that help the user to operate the meter more effectively.
Auto range AUTO 4. Page 19 Overload Alert The meter has provided an overload alert for voltage and current measurements. R2 command After executing R2 command, the meter will return the existing readings of secondary escorrt.
Parameters may be numbers, strings or boolean. It also explains the detail information of all RS interface command sets used in the meter. Moreover, a TCL session is created for handling the communication and the related data. Page 40 the LO limit value.
Table of Contents Add to my manuals Add. Page 17 Voltage, Resistance or Frequency Measurements Press the desired function escprt and connect the test leads as escrot in Figure to measure voltage, resistance, or frequency. Turn off the secondary display. Press then followed by to exit percentage function 3316a You must register and log in before leaving a comment. Selecting Trigger Mode once over the maximum display of Not used and always set to 0. Use of this instrument in a manner not specified herein may impair the protection provided by the equipment.
Releases the locking of the device’s keyboard. Plays an audio ‘beep’ checking that communication is ok mmA beep: Measuring AC and DC current on a rectification circuit. To avoid electrical shock or damaging the meter, never get water inside the case.
Page 12 Rear Panel The rear panel shown in Figure contains a line fuse, the power-line cord connector, an RS interface connector, and a cutout for IEEE interface optional connector. Since it is the default, the command can be shortened in: This display shows measuring readings, annunciator, and messages. The meter is acceptable only for the upper case part of command. It is presently not possible to handle more than 1 mmA multimeter per computer.
This command pauses program execution until all operations are completed. The meter will select the appropriate range in the auto-range mode, and an annunciator on the display will indicate measurement units. First scan the Tables of contents to be familiar with the outline of the manual.
Cancels 3136 “remote” mode.
Recommended Test Equipment Recommended Test Equipment The test equipment requirements listed in Table or equivalents are required to perform the calibration and performance verification test procedures.
Once the value of D5 has been set to 5 or -5, any adjustment to other digits will cause wscort setting value to subtract or automatic, respectively.
Page 34 Table Options And Accessories Options and Accessories At 33136a moment, one option is available for the meter, which option can be installed at the factory and a field installable retrofit kit option is also available: Page 20 Diode Continuity Tests Press to select diode continuity function, then connect the test leads across the diode under test as shown in Figure Reversing the polarity will reverse-bias the diode.
Resets the multimeter’s parameters to their default values. It also explains the detail information of all IEEE Sets the brightness of the displays.
For Windows, Linux or Unix. Add new file set. If RS of the meter is under print-only mode, the meter will print out the measured data when the measurement cycle is completed.